October 13, 2021
Ability Drive works on Tobii i-Series 13/16, Talk To Me eyespeak inspo and eyespeak alli, Tolt Technologies A13, and SmartBox GridPad 12.
On a Permobil chair you don't need an IOM if the chair has an Omnii 2. On a Quantum chair you don't need a SCIM if the chair has an enhanced display.
We haven't met an RNet system we don't work with. This covers Permobil, Sunrise, and Invacare Rovi. Quantum QLogic-2 (before 2017) systems have some safety issues and are not supported. QLogic-3 (Quantum Edge 2.0 and later) has no issues. Invacare models other than Rovi have not been tested.
In order for Ability Drive to work correctly the positioning of the tablet with eye tracker or AAC needs to be at 20-24" away from the face. The bottom of the tablet and eye tracker needs to be 6" below the chin. Tablet mounting position guide